The square sampling cup is for determining the Carbon equivalent after spheroidization, % Carbon, and % Silicon in molten iron by using the principle of thermal analysis.inquiry
The thermal analysis cup is used matching with the carbon and sulfur analyzer to quickly analyze carbon content, silicon content, and carbon equivalent in molten iron, which can save smelting time of molten iron, greatly reduce the production cycle of white and grey cast iron, and therefore increase production efficiency.
Materials: resin bonded sand
Temperature range: 1250℃～1370℃
Graduation No. of supporting thermocouple wires: K type
The shape of carbon cup: round and square type
Measurement function: thermal analysis towards cast iron
Measuring time: 240 seconds at most
Indication in the working process: the carbon cup is connected with three signal indicators on the panel of “Ready”, “Measurement” and “Finish” through potential free-reed relay.
One: option of the instrument and electric furnace
Be sure to select the specific carbon silicon analyzer and hot metal within the specified test furnace. the replacement of components when the next test furnace hot metal will be changed.
Two: data testing methods
Example: Thermal analysis of the test sample cup should be 3 to 5 consecutive (arrange it according to the time), e.g. the test result of Si% is 1.80%, 1.82%, 1.86%, 1.88%, 1.90%. Testing is completed record data.
According to the above testing result, the highest one is 1.80%, the lowest one is 1.70%, the fluctuation gap is 0.1, the average value is 1.75% (±0.05).
If the chemical analysis of Si% molten iron content was 1.85% when the original apparatus should be set based on the si values of 0.1 and then raised to complete the adjustment (if the former setting of Si is -0,2, should change at -0.1). when tested again, the content of Si% will become 1.85%, the adjustment is over, can work now.
1. The hot metal temperature in the sample cup should not exceed 1350 ℃.
2. Optimum temperature of ductile iron is 1200 ℃ ~ 1250 ℃.
3. Optimum temperature of Gray Iron 250, 300 is 1250 ℃ ~ 1350 ℃
4. The abnormal situation like low carbon high silicon hot metal composition will cause instruments to measure the normal C% and Si% no data.
First of all, our factory adheres to the principle of quality first. We have a group of professional technical and inspection personnel to focus on product quality control and Technological innovation. For each product, we have strict process inspection and factory inspection. We have a well export experience, to provide customers with professional pre-sales and after-sales service and technical support. The sales process will be active tracking queries, positively solving the problems in the follow-up use of customers.